Potentially harmful product represented as “Generic Tamiflu”- tests revealed product does not contain oseltamivir, but cloxacillin, an antibiotic.
Master Bond’s Polymer Adhesive Supreme 10AOHT-LO is a single component, no mix, thermally conductive compound featuring exceptional electrical insulation and toughness. It meets NASA low outgassing specifications and has
Some liquid Vitamin D supplement products are sold with droppers that could allow parents to accidentally give harmful amounts of Vitamin D to their infant.
The Kontron Gigabit Ethernet Switch VX3910 offers 3U VPX (VITA 46.x) and OpenVPX (VITA 65) platforms Enterprise-Class Switching functionality with a total of 28 Gigabit Ethernet ports and advanced management features. With its
FDA evaluating data in which type 2 diabetes patients taking Benicar had a higher rate of death from cardiovascular cause compared to placebo. FDA has not concluded that Benicar increases the risk of death.
Injectable products may contain particulate matter.
AED may falsely detect an error condition during charging for a shock, then cancel charge and not provide therapy, when used with an affected battery pack.
Selco Products' CP Series and CS Series Thermistor Probe Assemblies can be customized to meet unique demands for specialized application requirements. These negative (NTC) temperature coefficient thermistors can be
Heyco Products has announced that its Snap-In Ratcheting Strain Relief (PN 1280) is now removable. The newly designed strain relief can be easily removed through insertion of standard snap ring pliers with
Tumbler Technologies + TRUMPower introduced its medically approved TMC200 Series power supply. It has a compact size of 3” x 5” x 1.5” and is capable of delivering 200 W of continuous DC power. The series is
Market withdrawal due to an increased number of adverse event reports of allergic reactions associated with two lots.
The four products were manufactured at a plant cited by FDA for deficiencies in Good Manufacturing Practices. Posted 05/30/2010
Products sold under the Claris, Sagent Pharmaceuticals, Pfizer, and West-Ward Pharmaceuticals labels. Reports of floating matter in intravenous bags of metronidazole and ondansetron. Foreign matter should not be present in a sterile injectable product.
North Atlantic Industries (NAI) has announced the release of the SIU6 Sensor Interface Unit, a rugged, packaged, turn-key Embedded I/O system for remote data acquisition and distribution for large platform monitoring and control system applications. This high functional density system features
Tech9 Corporation’s 1 A buck-boost DC to DC power converter accepts input ranging from 6 V to 70 V while output remains constant. The power converter can be ordered in 3 V, 5 V, 9 V, 12 V and 15 V output models. These power converters are
Product contains substances which are poisonous when ingested or applied to large areas of the body.
Voluntary recall of all manufactured lots has been initiated as a precautionary measure due to instances of particulate matter observed in a limited number of vials during routine stability testing.
JDSU announced that it has created new frameless display panels using a combination of its low-reflection optical coating technology and custom display manufacturing processes. The result are
Revised labeling to include cases of severe liver injury.
Fujitsu Components America, Inc., purportedly the first manufacturer to obtain Windows 7 logo certification for a resistive multi-input touch panel, is marketing those touch panels in North America. Using Fujitsu’s proprietary technology, the resistive multi-input panels can
Possible increased risk of fractures of the hip, wrist, and spine.
Risk of overdosage or suicide for patients who are addiction-prone, taking tranquilizers or antidepressant drugs.
Stackpole Electronics expanded its NMCR and NMCF Series of non-magnetic thick film chip resistors to include all EIA standard chip sizes from 0402 to 2512. This expansion allows
Non-sterility of metronidazole injection administered via the intravenous route has the potential to result in infections, which could be fatal.
Defective control board wiring harnesses can cause the machine to unexpectedly shut down, terminating ventilation, anesthetic delivery, and potentially patient monitoring.